The aim of this booklet is to help the reader to: Appreciate the basics of electrical safety. Understand the reasons behind and the purpose of the IEC publication. Provide an understanding of the benefits of using the different tests available, in order to help them prepare the adoption of the IEC standard. Please complete the form on this page to access the download link for our free guide to IEC Introduction This section talks about electrical current, the human body and how it conducts current, and IEC Body Model.
|Published (Last):||23 March 2019|
|PDF File Size:||3.40 Mb|
|ePub File Size:||11.93 Mb|
|Price:||Free* [*Free Regsitration Required]|
Medical electrical equipment - Recurrent test and test after repair of medical electrical equipment Abstract IEC applies to testing of medical electrical equipment and medical electrical systems, hereafter referred to as ME equipment and ME systems, or parts of such equipment or systems, which comply with IEC second edition and its amendments and IEC third edition and its amendments, before putting into service, during maintenance, inspection, servicing and after repair or on occasion of recurrent tests to assess the safety of such ME equipment or ME systems or parts thereof.
For equipment not built to IEC these requirements may be used taking into account the safety standards for the design and information in the instructions for use of that equipment.
This standard contains tables with allowable values relating to different editions of IEC For the purpose of this standard, the application of measuring methods is independent of the edition according to which the ME equipment or ME system is designed. This standard contains "general requirements", which contain clauses of general concern, and "particular requirements", further clauses handling special types of ME equipment or ME systems and applying in connection with the "General requirements".
This standard is not suitable to assess whether ME equipment or ME systems or any other equipment comply with the relevant standards for their design. This standard is not applicable to the assembly of ME systems. This standard does not define requirements for repair, exchange of components and modification of ME equipment or ME systems.
Otherwise conformity to applicable requirements should be assessed and verified, before the tests of this standard are performed. This standard is also applicable to tests after repair. This second edition cancels and replaces the first edition of IEC published in This edition constitutes a technical revision including the following main revisions: - clarification in 5.
Free Guide to IEC 62353
IEC System of Conformity Assessment Schemes